If your instrument does not provide any more remote power supply to the LNB or multiswitch, check the following points.
a) Check if the switch “DC@RF IN”, available in most models, is set to ON: see examples figures 1 and 2.
|Figure 1: DC@RF IN switch example||Figure 2: DC@RF IN switch example|
b) Ensure that the YELLOW LED “DC@RF IN” on the front panel of the instrument lights on;
c) If the LED lights up, but after a few seconds that you have connected the LNB/Multiswitch especially if using LNB SCR or dCSS with a high consumption (150/200 mA), the instrument does NOT provide any more voltage (the spectrum of the received signals decreases sharply or measures are interrupted), check:
In this case, the instrument must be sent to ROVER for repair (NOT covered by warranty).
|Figure 3: PTC example (60 Volt, 650 mA, 0,31 Ohm)|
IMPORTANT: For safety reasons, the power supply voltage, SWITCH OFF automatically when changing function, for example by pressing the HOME key or when power is turned OFF.