Below we listed the procedures to perform POWER AND OPTICAL ATTENUATION MEASUREMENTS:
|Touch “OPTIC” from the HOME screen;||Touch “WAVELENGTH” and select the
required wavelength, for example “1310nm”;
|Touch “DC@RF” and select, if it is required,
the power supply voltage: for example “12V” (Volt);
|Touch “STORE” and store the measured optical
power value (OPTIC REF.): for example “6,07 dBm”;
|The “OPTIC POWER LOSS” field shows the optical attenuation value
compared to the stored value (Optic REF): for example: “-0,53 dBm”
(OPTIC REF. 6.07dBm less OPTIC POWER 5.54dBm uqual OPTIC POWER LOSS -0.53dBm).
Below we listed the procedures to perform OPTICAL INPUT RF MEASUREMENTs & SPECTRUM:
|Touch “RF IN” and select “OPTICAL” from the volume screen;||In SAT mode press the“PLAN ” key,
select the satellite required,
then “SPECT” to visualize the spectrum;
|SAT spectrum;||Press the “SAT” or “MEAS” key to carry out the measurements
(depending on the model of instrument used).
FIBER OPTIC and REMOTE POWER SUPPLY CABLE CONNECTION example:
Important: In OPTIC model it is possible to analyze the spectrum and measure only Vertical/Low band (VL) transponders.